Ventilator, Intensive Care, for adult and pediatric patients, AC and battery powered, with accessories
Oxygen concentrators are not suitable for use as an oxygen source for mechanical ventilators. The technical specifications listed below are the minimum requirements that ventilators can perform and fulfill according to the WHO requirements. The detailed technical specifications and contents for each ventilator model may be slightly different. Therefore information relating to the specific product model i.e. product brochure, package insert, etc. to be supplied will be sent to the Principal Recipient for acceptance and approval prior to ordering
Note: Ventilators are complex devices which require trained medical staff to operate and qualified technical staff for maintenance. In addition, ventilators require specific infrastructure provisions to be able to operate. Therefore, ventilators should only be ordered after consultation and necessary approvals with in the Ministry of Health
At least IP21 degree of protection to the harmful ingress of water (fluid spill resistance)
FiO2: 21 to 100 %.
Minimum Inspiratory flow: 1 – 160 [L/min]
Minimum Inspiratory pressure: 0 – 40 [cmH2O]
Minimum RR: 10 to 60 [breaths/min]
Minimum PEEP: 0 to 20 [cmH2O]
Minimum available ventilation mode: A/C-PC Assist control (pressure control) ; A/C-VC: Assist control (volume control); CPAP: Continuous positive airway pressure; PSV: Pressure support ventilation; S/T: Spontaneous/timed ventilation; SIMV-PC: Synchronized intermittent mandatory ventilation (pressure control); SIMV-VC: Synchronized intermittent mandatory ventilation (volume control)
The availability of alarms, related to gas delivered; High/low FiO2;High/low inspiratory pressure and PEEP; High/low tidal volume (not achieved or exceeded); Apnoea, adjustable from 10-30 sec; High/low respiratory rate; Continuously high pressure/occlusion; Breathing circuit disconnect.
The availability of alarms, related to equipment operation: Visual and audible for: Gas supply failure; Power failure; Low battery.
Ventilator main unit – 1 piece
Power cable – 1 piece
Filter – 1 piece
Patient circuit – 1 piece
Humidifier – 1 piece
Stand – 1 piece
User Manual – 1 piece
Quality Certificate – 1 piece
Operates at temperature of 5 to 40 °C and humidity of 0 to 85% RH
Operates from AC power electric line: 100-240 V , 50/60 Hz.
In-built rechargeable battery. Automatic switch from AC power electric-line mode to battery operating mode and vice versa. Equipment must be connected to a continuous source of power. Note that power plugs may vary across countries