Non Invasive Ventilator, Bi-level positive airway pressure, for adult and pediatric mode, AC powered, with accessories
Oxygen concentrators are not suitable to be used as an oxygen source for other medical equipment providing respiratory support, such as ventilators, CPAP devices, High Flow Nasal Cannula, etc.
The technical specifications listed below are the minimal requirement that the BiPAP devices can perform and fulfill according to the WHO requirements. The detailed technical specifications and packing contents can be slightly different across models. The specific product model and information will be sent for your approval prior to next steps.
Ventilation Mode: S/T, CPAP, S, T, PC
Pressure Range : 4- 30 cmH2O
Connectable to oxygen source
Alarms, related to gas delivered; Visual and audible for: High/Low Temperature; Breathing circuit disconnection.
Availability of alarms, related to equipment operation: Visual, audible and clearly indicating the problem for: Lack of water; System failure; Power failure; Low battery.
Operates from AC power electric line: 100-240 V ~, 50/60 Hz.
Automatic switch from AC power electric-line mode to battery operating mode and vice versa. Equipment must be connected to a continuous source of power. Note that power plugs may vary across countries
BiPAP Main Unit – 1 piece
Breathing Tube – 1 piece
Humidifier Tube – 1 piece
Full face Mask – 1 piece
Trolley – 1 piece
Humidifier – 1 piece
User Manual – 1 piece
Power cord – 1 piece
Quality certificate – 1 piece
The User Manual supplied with the device details all installation which can be undertaken by users.
The User Manual supplied with the device details all operation procedures for user.
The User Manual supplied with the device details all maintenance and troubleshooting procedures which can be undertaken by users.
Pozzallo Note: Contact Iplussolutions if additional installation and training requirements are required beyond user manual